Question about Braun Health & Beauty
Ive the same problem. it has new batteries and been cleaned but still reads lo or a low temperature. why?
Posted on Apr 26, 2012
Temperature taken is not within typical human temperature range (93.2 °F - 108 °F / 34 °C - 42.2 °C).
Make sure new, clean lens filter is properly inserted. Then, take a new temperature.
Posted on Mar 17, 2011
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
May 19, 2018 | Braun Health & Beauty
Mar 18, 2018 | Cars & Trucks
Sep 22, 2016 | Braun Irt 4520 Thermoscan Ear Thermometer
May 15, 2014 | Home
Jan 08, 2014 | Braun Irt 4520 Thermoscan Ear Thermometer
Mar 14, 2013 | Braun Health & Beauty
Jul 03, 2012 | Braun IRT3020 Thermoscan Ear Thermometer
Jun 25, 2017 | Health & Beauty
Nov 11, 2011 | Braun Health & Beauty
Feb 28, 2011 | Braun Health & Beauty
2,095 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: