Question about Health & Beauty
Calibration is required, I hope this helps, regards Jem ????
Posted on Sep 22, 2019
The C error message can indicate a problem with the weight measurement or a need to recalibrate the scale. Make sure that the scale is on a hard flat surface. Try stepping off the scale and back on. Make sure to stand still on the scale. If the problem continues, step onto the scale with one foot until the display lights up. Step off the scale and wait. The display should read 0.0 if the unit has been recalibrated.
Several Etekcity scale manuals are available here: https://www.etekcity.com/support/manual?_url=%2Fsupport%2Fmedia . The calibration step is usually required if the scale has been moved or turned upside down.
I hope this helps.
Posted on Sep 22, 2019
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
EACH BUTTON IS FOR FOUR INDIVIDUAL USER. BUTTON ALLOWS YOU TO INPUT IDEAL WEIGHT (FOUR GOAL WEIGHTS FOR FOUR INDIVIDUAL USERS). AND CALCULATE AGAINST ONE OF THE FOUR INDIVIDUAL ACTUAL WEGHTS. i AM USER 1 IN MY FAMILY. I HIT THE 1 BUTTON, WEIGHT AND MY GOAL WEIGHT COMES UP BEFORE MY ACTUAL.
Posted on Mar 08, 2009
I found this from another posting. Hope it helps.
FEMALES: age 20-29 normal= 6.2-7%
FEMALES: age 30-39 normal= 6-6.8%
FEMALES: age 40-49 normal= 5.6-6.4%
FEMALES: age 50-59 normal= 5.2-6%
FEMALES: age 60-69 normal= 4.9-5.7%
sorry no listings over age 69
My only thing now, is what it means if I am below the range.
Posted on Jan 13, 2010
Take a 5 pound bag of sugar, flour or what have you. Put it on the scale. If it is off, turn the scale so that you can check every possible place for an adjustment. Adjust and go for it!!! Good luck...Lee
Posted on Jan 17, 2010
The zero capture is out of the 2percent tolerance so the cell is either faulty or requires zeroing. Open the unit and press the micros itch on the left of the board. The display flashes s. Press the zero key to change to flashing E. Press store to scroll to zero balance. Press semi auto tare. This captures zero without clearing the span calibration. The display will scroll around for a short period when complete press store.. Release the micro switch and scale will go thru warm up routine then display zero. Hope this helps, regs Darren.
Posted on Apr 15, 2012
Tips for a great answer:
Nov 07, 2017 | Health & Beauty
Sep 15, 2014 | Health & Beauty
Aug 20, 2014 | Office Equipment & Supplies
Oct 14, 2012 | Health & Beauty
May 16, 2010 | Microwave Ovens
Mar 12, 2010 | Tanita Health & Beauty
Jan 16, 2010 | Jobar International Idea Works Wireless...
Jun 15, 2009 | Balance Health & Beauty
Jan 26, 2009 | Salter Stainless Steel Electronic Bathroom...
29 people viewed this question
Usually answered in minutes!