Question about Dewalt DW077KD T Self Leveling Rotary Laser Combo Kit With Free Tripodconsist Of DW077KD And DW0736 P...
I am laser technician, i have a DW073 out of calibration. I need instructions to calibrate. I have a colimator, and laser is off the screen.
Sounds like you figured this out on your own. I calibrate my own DW073 regularly without any fancy gadgets. I use the calibration check procedure in which the unit is placed about midway between fixed objects 100' apart.
These have a front and back and sides, so first you mark where the laser hits facing front to back at your targets. Then turn it 180 deg. relevel and resite. If the difference between the new and old marks is not the same then I turn the adjuster to halve the inconsistency and once that is accomplished, tweek the adjusting screw at the end of the level vial. (Because if you have adjusted more than the tiniest bit, the bubble will no longer display as level. So having established level, or closer to it, the adjustment here is actually to bring the relevant bubble vial back to level rather than assume they are level and adjust the physical beam mechanism (unlike the old physical sight levels where you adjusted the actual optics -- a little more like adjusting a rifle scope). Then you repeat for the side to side vial.
Anyway sounds like you are set so I posted this for anyone who happens to find this thread and is looking to field calibrate their DW073, e.g. don't have time or money to get it done in some shop.
I also wanted to post an add on question. Did you figure out how to adjust the "cone error", i.e. the accuracy of alignment of the twin lasers. They appear to have allen set screws that lock their adjustment, and maybe they are then reaimed by using a slotted screwdriver on the brass ring surrounding the laser. Maybe these are made eccentrically or slightly beveled so turning the laser reaims.
So far, my cone has stayed within the specs of 3/32" at 25' in the manual. But I'm seeing two lines at maybe 150' out. Probably that is close enough for gvmt. work, and I may find the adjusment so sensitive that I can't really get it any closer. But I'd at least be interested to know the method for adjusting the cone error.
Posted on May 06, 2010
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Jan 09, 2013 | Drills
Jan 04, 2012 | Office Equipment & Supplies
Apr 01, 2011 | Dewalt DW077KD T Self Leveling Rotary...
Jan 31, 2011 | Dewalt & Heavy Duty Rotary Laser Kit
Oct 09, 2010 | Dewalt Dw087k Selfleveling Line Laser...
Jun 17, 2010 | Stanley 77 156 RL100 Manual Leveling Dual...
Nov 27, 2009 | Dewalt DW077KD T Self Leveling Rotary...
Jul 13, 2009 | Tools & Hardware - Others
3,682 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: